ASTM D is a significantly updated version of the Standard Test Method for Determination of Thermal Conductivity of. Soils and Rock by Thermal Needle . ASTM D – Standard Test Method for Determination of Thermal Conductivity of Soil and Soft Rock by Thermal Needle Probe Procedure. This standard is. Buy ASTM D TEST METHOD FOR DETERMINATION OF THERMAL CONDUCTIVITY OF SOIL AND SOFT ROCK BY THERMAL.
|Published (Last):||13 August 2007|
|PDF File Size:||5.12 Mb|
|ePub File Size:||7.73 Mb|
|Price:||Free* [*Free Regsitration Required]|
TK04 can use both methods by simply changing the connected probe. Nails, Spikes, and Staples. It is recognized also that it would be unwise, because of the standardization of this test method, to resist in any way the further development of improved or new methods adtm procedures by research workers.
TP08 – Hukseflux USA
Chosing an evaluation method is a compromise: Today, some 12, ASTM standards are used around the world to improve product quality, enhance safety, facilitate market access and trade, and build consumer confidence. Permanent installation in soils: It can be used where standard size probes are too large for the available sample dimensions. For permanent installation in soils, a dedicated model, TP01, is available.
A separate brochure is available. When using the modified method, part of the heat generated during measuring penetrates into the probe body instead of the sample material.
Usually this is achieved by clamping the base into the same material metal that holds the sample. Ast thermal conductivity of the sample is calculated from the temperature rise with time registered by the probe. A high quality measurement with TP08 requires that not only the sample but also the base are at a stable preferably the same temperature.
In general a NSSP consists of a heating wire, representing a perfect line source, and a temperature sensor capable of measuring the temperature at this source. In some cases samples can even be smaller consult Hukseflux.
Reliable testing depends on many factors; Practice D provides a means of evaluating some of those factors. TP08 is a probe that offers the possibility to perform a practical and fast measurement of the thermal conductivity or thermal resistivity of the medium in which it is inserted at a high accuracy level.
They yield more accurate results, and they can be used to detect disturbances in the data by ashm the measured temperature curve to the theoretical curve form. The slower the source temperature rises, the higher is the thermal conductivity of the sample material.
Downloads Download Product Manual. The NSSP principle relies on a unique property of a line source: A correction method developed by TeKa uses the thermal parameters of probe and sample to automatically determine the effective amount of heat entering into the sample material.
Agencies that meet the criteria of Practice D are generally considered capable of competent and objective testing. The inch-pound units given in parentheses are for information only. This test method is suitable only for isotropic materials.
The d5334-08 common approximation method calculates thermal conductivity fron the slope of a straight line fitted to the measuring data. Min 20 mm radius.
Active Risk Management
This ensures consistent results for the conventional and the modified needle probe method in the complete measuring range. Register for a trial account. Enter your personal account email address to request a password reset: This improves reliability significantly, because thermal conductivity measurements are very sensitive to sample preparation problems and instable measuring conditions.
The sample size is not critical, as long as a 20 mm radius around the needle is covered. The standard TP08 probe has proven its suitability for soils, thermal backfill materials, sediments, foodstuff, powders, sludges, paints, glues and various wstm materials.
Active view current version of d334-08. In TP08 the reference junction of the thermocouple is located in the base.
Are you sure you want to empty the cart?